Source: American Environmental International, Inc. (AEI)
The Model 502 utilizes a flame ionization detection technique for the continuous measurement of total hydrocarbons, as equivalent methane
The Model 502 utilizes a flame ionization detection technique for the continuous measurement of total hydrocarbons, as equivalent methane. The sample gas is mixed with a fuel (normally a composition of hydrogen and nitrogen) and burned in an atmosphere of blanket air. Ions formed in the burning process cause an electrical conduction between two electrodes in the combustion chamber that is amplified by a high-sensitivity electrometer-amplifier. The electrical output of the electrometer-amplifier is linearly proportional to the quantity of flame ionizable hydrocarbons present in the sample stream. The signal is linear over a range of 0 to 10,000 ppm.
- Air pollution...measuring hydrocarbon emissions from industrial processes emissions.
- Quantifying contaminants in ambient air or plant air.
- Monitoring for fuel leakage or toxic solvents.
- Detecting contaminants in vent gases from exhaust hoods and drying ovens.
- Monitoring combustion efficiency and monitor LEL concentrations in process streams.
Typical Industrial Applications:
- Printing Plants
- Painting Operations
- Pulp and Paper Plants
- Oven Process Drying Applications
- Textile Manufacturing
- Can Coating Operations
- Pharmaceutical Plants
- Chemical Process Plants
American Environmental International, 725 West University Drive, Arlington Heights, IL 60004 Phone: 800-343-8601; Fax: 847-342-8500